A frequency sweep from F1 to Fn then obtain 2 different output such as C and R for each frequency using Agilent E driver vi’s. Input data which is a 2D array is ok but in output which is a string the middle data are zero!! Message 3 of 8. Thank you for your consideration and time dear Dale. I can not find the reason. Thank you in advance.
|Date Added:||24 July 2013|
|File Size:||35.26 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Thank you for your reply. The array consists of numbers and text like: You should be able to find it here: If you place this after the Initialize vi I think it should work.
What kinds of instrument drivers support the E4980A?
Message 3 of 8. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Message 6 of 8. The example seems to gather data points between Hz and Hz, then data points between Hz and Hz and then data points between Hz and Hz. I found out that the problem is in labiew of e44980a “array to spreadsheet string”vi.
Message 8 of 8. I think I almost finished the code.
It sweeps between 3 frequencies and has high and low limit bands labviwe I dont know what these bands are. I connected the device to the labview. The array consists of numbers and text like:. Thank you in advance. It does not give the data.
need help: Agilent EA LCR meter through GPI | Keysight Community
Yes I have tried that example. After finalizing it I will publish it here for those who one day may need it.
I want to measure capacitance and resistance in a frequency domain. I wrote a code which uses single mesurment agilet in a FOR loop which adds a specific amount to frequency amount and then again conducts another single mesurment and gave the results at the end of loop.
Driver EA(agilent) for Labview
I used probes for debugging. In that example the sweep is between 3 points. Most Active Hardware Boards: Thank you for your consideration and time dear Dale. Message 5 of 8. I would like to verify that this is correct, but I do not actually have the hardware available to me.
Keysight Instrument Drivers
Message 1 of 8. Once you have the data collected and stored as an array perhaps, you can write to a spreadsheet with one of the VIs mentioned here: As for choosing measurement type, I took a look at the other included example and saw that it uses the Configure Measurement vi to select a Function one of them being Cp-Rp. Input data which is a 2D array is ok but in output which is a string the middle data are zero!!
Do you know how I should modify this example? Hello, I want to measure capacitance and resistance in a frequency domain. I am using keysight driver Labview package instead of Visa.
But i have problems with input and output of these vi’s. It doesnt have the outputs. The maximum number of measued data is in this LCR meter.