Once you have completed these steps, you need a fast way to progress from interactive mode to programming mode so you can immediately begin writing your tests without a tedious transition process. You can also find these at ni. If it is not, then you know to check cables, confirm power, and verify that the instrument is working properly. If you have two or more instruments on the bus, you can disconnect all instruments except one to determine its address. You also can use the Resource to Find field to query each instrument for an identification ID string.
|Date Added:||12 April 2007|
|File Size:||41.26 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Because most instruments are compliant with Bob Schor 28 August You can repeat this process to confirm communication with all of your connected instruments.
As I understood this is your case, yes? If there is one, then uninstall the Keithley driver then load the NI driver.
Keithley 2260b-gpib-usb GPIB to USB Adapter
Once you have completed these steps, you need a fast way to progress from interactive mode to programming mode so you can immediately begin writing your tests without a tedious transition process. With this test panel, you can uusb properties for your instrument communication as well as read and write to the instrument.
Message 1 of But I don’t know HOW? I did as you said.
If we cannot see “what you did wrong”, we are unable to tell you how to fix it. To communicate with our instrument at Address 4, follow these instructions:. Back to top Related Portals: Otherwise, the GPIB interface is detected when you start your computer.
ID querying is discussed in the next section. When VISAIC initially runs, it automatically finds all of the available resources in the system gpkb lists the instrument descriptors for each resource under the appropriate resource type.
Instruments typically respond with the manufacturer’s name, model name, and various alphanumeric characters that gpub manufacturer uses to track firmware revisions.
Installing GPIB-USB Controllers – National Instruments
If your system involves different ground potentials, the voltage difference could surge through the GPIB hardware and cause damage. The top process industries grew at a faster rate than the top discrete industries inand this trend continued inaccording to the Will you be kekthley your Keithley Instrument with their software?
My Cart Books Online Training. If your computer is already running, the operating system automatically detects the GPIB interface.
Message 9 of Unfortunately, engineers sometimes forget to reconnect all the cables and power up all the instruments. I have attached the screenshot If there is a problem, you no longer see your device. Electromate introduces Lintech series linear positioning slides The linear profile rails are mounted to a machined aluminum base, which offers support over the entire travel of the table’s carriage.
Or at least you have a manual describing the command protocol?
This is especially important for large systems featuring many instruments. Alternatively, system power may be disrupted for a variety of reasons or the actual instrument may begin to malfunction. The best way to achieve this is by using instrument drivers. Hmm, could keitgley explain this further?